1) Do test runs to establish
optimum settings to detect Triton.
Test runs are best done on nights prior to
that of the occultation, as well as during the run-up to the occultation
itself. The star is brighter than
Triton, so it is worth while to give the event a try
2) The planet is 100x brighter, so Neptune will create unwanted light:
CCDs often produce leakage along the pixel
columns (and sometimes along the row direction), so it is sometimes
beneficial to turn the camera to minimise overlap of stray light interfering
with the image of Triton, rather than having it aligned in RA and DEC.
The WinJupos plot (above) indicates Triton is at PA (about) 110 degrees.
3) Will Moonlight effect
visibility ? The event happens just 5 hours
after the Moon is full and quite close to perigee, and higher in the
sky than Neptune. With an angular separation of 34 degrees, it may be
worth observers erecting a temporary 'moon shade' to stop moonlight
entering the telescope aperture. It might mean the difference between
success or failure for observers using relatively small apertures.
4) A telescope with long
focal length will produce best results. e.g. SCT: There
are fewer diffraction effects, and contrast should improve, so a SCT
would be a good system to use.
5) Is use of a Barlow lens
is indicated ? The Neptune / Star separation
is 10" arc so a Barlow will increase the image scale. Getting
good focus on the star will also be important. It may not be easy in
poorer seeing, but if the star is recorded then we may get a result.
Use of a Barlow should be part of the test.
6) Take as many short exposures
as possible. Subsecond exposures between 0.16
to 0.64 sec are desirable depending on sky conditions and the aperture
7) Aim for good sensitivity.
i.e. Triton+Star are detected clearly. If
necessary longer exposures should be used e.g. 1sec in order to capture
8) If possible, use a 12-16bit FITS, otherwise 8bit AVI: Part
of the extended test. This observer will use 8bit analogue video to
(AVI) to start with.
9) Use of a USB planetary web cam requires the clock be Sync 'ed with
Dimension4 (Thinking Man Software) or similar for NTP. Correct
computer time is important.
10) Suggested software is: FireCapture or SharpCap for time stamping
(or use an established method): Planetary
web cams are useful, but the user will need to establish their sensitivity
using a trial run.
11) It's not thought that
filters will improve detection for this event:
Since the planet is blue/green, the option would be a red filter. But
since the spectral type of the star is not given, filtration may not
provide any benefit.
to filtration: Atmospheric Dispersion will
also reduce the amount of focused light reaching the detector. There
is a good description of the ADC here.
I am tempted to try a light green W11 filter during the test phase.
Filters). The use of an Atmospheric
Dispersion Corrector might improve the detection limit by sharpening
1 - RESULTS FROM TEST VIDEO-ING
ON 2017 JULY 21 @ 0227UT
to the above image:
This was a 30cm F4 and 5x Powermate, with WAT-910HX Setting x32, 39dB
The camera was pre-focused with a Bahtinov mask using alpha And. Stability
was Ant iii (not good, not bad either).
A single integrated frame (0.64s) was selected for this illustration
( H. Pavlov)
- RESULTS of TEST with a University KLEE 2.8 x Barlow
2017 July 26, 0200UT
compared to NO Barlow (Filtered with Wratten 11)
used for the image below: Native focal length is 1200mm @ F/4
Notes to the above test-2
By comparison with the first test, the KLEE Barlow  gave about the
same amplification as a 5 x Powermate [ FL 6m ]. This was explained by
the fact the WATEC CCD with nose piece and filter was positioned 35 mm
from the normal eyepiece focal plain. This increased the amplification
to about 4.5 (or 5). From an approximate calculation  the focal length
of the KLEE was 25mm (1 inch). Much shorter than most
other Barlows. With this in mind I measured a Revelation Astro ED 2x Barlow
(2" fit) and found this to have a focal length of 100mm.
 Loaned by Adrian Jones (Maidenhead AS)
 J. B. Sidgwick, Amateur Astronomer's Handbook. 2nd ed 1960
Tangra Light Curve from the
using 16/50s, gain 34dB NO Barlow, W 11 filter, with "pixels in
the background annulus" reduced from 350 to 100 (Tangra>settings>photometry).
- RESULTS of TEST with an 2x ED-Barlow on 2017 July 28, 0200UT
with Wratten 11 (pale yellow-green)
Notes to the above
Smaller angular separation from Planet (10-12") due to orbit. Scale
is 0.7 arcsec / pixel. Wider field of view ( 7' x 9' ) so periodic error
is less obvious. A single field star of Mag 13.5 was available as a
guide star for the photometric software. Seeing stability was poorer
compared to previous test images
- RESULTS of TEST with a 1.25" KLEE x2.8 Barlow on
2017 Aug 12, 0030UT
with and without Wratten 11 (pale yellow-green).
In Moon light [89% 15d to East]
Notes: Triton is magnitude 13.5. The star UCAC4-413-143127 is 1/3rd
mag fainter than the occulted star (Not in the image). This is an
unfiltered CCD video in transparent conditions (no cloud or haze)
but seeing stability was poor. I estimated the seeing disk from 2
to 4" arc variation. The effect of the Moon nearby at 15deg.
distant and similar altitude is now causing a noticeable increase
in background noise event at F12 effective focal ratio. Even so the
integrated frame of 0.64s clearly shows Triton above noise level.
1)Signal to Noise unfiltered.........2) Target star (mag 12.4).........3)Effect
of adding theW11 filter
Notes: Signal to noise was estimated with TANGRA. Three
screen captures are compared.
(1) Unfiltered: An area of blank sky (green line) was compared
to Triton (Blue line). The StN was 9 and the uncalibrated mean intensity
is about 1700
(2) Unfiltered: An area of blank sky (yellow) was compared
to the predicted target star. The StN was 25 and the uncalibrated
mean intensity is about 2500
(2) Wratten 11: Comparison
of Triton (blue) with field stars. The StN on Triton had dropped to
about 5, and the uncalibrated mean intensity is about 1300.
The W11 filter subtracts too much light with no added benefit (that
could be seen). The presence of Moon light suggested that the best
StN is required (i.e No Filter), so no filter will be used.
See the New Observers panel on http://www.euraster.net/
Or download it from http://www.euraster.net/results/report-form.txt
Please report the observation
to the BAA Asteroids and Remote Planets Section for preliminary analysis
and observer feedback.